Just like all machines daily centrifuge maintenance and cleaning is needed for the tool to use properly. A few steps are expected to ensure the centrifuge runs in optimal performance with minimal centrifuge repairs with little or no human supervision as possible.
Daily cleaning of the inside surfaces of the centrifuge with a dilution of household bleach or an equivalent disinfectant is crucial. Portions of the centrifuge in contact with blood or other potentially infectious agent must be immediately cleaned when tube cracks or breakage ever happens.
All loose materials like plastic or broken glass must be carefully removed and disposed. Measured and rated speeds should not differ by significantly more than five percent under specified conditions. Centrifuge speeds should periodically be checked utilizing a reliable photoelectric tachometer in respect with the College of American Pathologist (CAP) inspection guidelines. Proper centrifuge maintenance and centrifuge repair guidelines requires that all centrifuge bowls should be cleaned with an antibacterial solution. Rotor heads and buckets should be cleaned as well.
Using CAP guidelines , the accuracy of the centrifuge timer should also be checked and verified on a regular basis. Measured and programmed temperature should be within two degrees Celsius. The temperature of refrigerated centrifuges should be checked at least once a regular in standardized conditions Netapp Support If one does not keep to this, then centrifuge repairs of machine parts may be required. Centrifuges with seven hundred to fourteen hundred running hours after minor repairs should be carried out, good maintenance and repair of centrifuges is a significant guarantee for its safe operation. Machines running between thee thousand to four thousand work hours will require an overhaul with repairs to any damaged drums, corroded shafts, rim fractures, and loose shells. Minor repairs must be performed to wash the bearing components and tighten all bolts. When doing centrifuge repairs , replace the screen, brake band, check the bearings, and replace the rubber shock absorbers.
Manufacturer’s instructions for lubrication, maintenance and replacement of brushes should be followed on a regular basis. Failure to replace worn brushes might cause the motor to fail and require replacement. All centrifuge maintenance function checks must be recorded and all corrective actions documented. A preventative maintenance schedule is the better means for customers to take care of their equipment, without the headache of getting to remember to schedule service. It’s automatic, and you can choose the month of the year that is most effective for you!Other